Reliable Test Equipment Manufacturer
China Top Ten Brands Of Test Instrument
Time：2019-09-12 Tag：Technical solution User：Sailham View：292
Constant temperature and humidity test chamber: treatment method when the humidity cannot fall down, let see here
1. Confirm whether the wet & dry ball gauze is hung on the sensor, and the position of the half of the gauze should be in the sink (the gauze needs to be replaced regularly, and it should not be stained with oil or foreign matter, keep it clean).
2. On the right side of the chassis, check if the water level of the water cup is too high or too low.
3. When the equipment is running, judge whether the pump is in working condition to check whether there is vibration or not.
4. Check whether the humidification tube is abnormal.
For more information, contact us sailham.com, climate test chamber manufaturer.